Footwear Tester for Individual Measurement & Evaluation of Electrical Resistance
Simple, yet user-friendly
Very easy-to-use footwear tester. Information of measurement and evaluation of electrical resistance at a glance.
Meet the criteria of standards such as JIS T 8103:2010 and ANSI/ESD S20.20.
Measure the anti-static performance of shoes individually. Prevents faulty measurement with the divider on test plate.
Evaluation external output
Evaluation results can be output through the external output terminal to interlock with other device, to open door, turn on signal, etc.
・When PASS LED is lit
It indicates that the resistance of the anti-static shoes and the person wearing them is inside
the set range.
The resistance range of the shoes at that time is indicated by the LED lamps,
therefore the characteristics of the anti-static shoes can be checked.
・When FAIL LED is lit
It indicates that the resistance of the anti-static shoes and the person wearing them is outside the
Determine the cause and take actions including replacing the anti-static shoes, etc.
If the measured value is below the lower limit
Example) Soles are worn down. A piece of metal is stuck in the sole.
※If the measured value is above the upper limit
Example) Thick socks are worn. Defective shoes, dirty soles, etc.
DIP switches for various settings
Upper and lower evaluation limits, evaluation buzzer, etc. can be made by setting the DIP switches.
Any impact on people with heart conditions, people using cardiac
pacemakers or pregnant women?
A small amount of electric current will flow through the human body when using HAKKO FG-460.
Please do not use HAKKO FG-460 if having a heart condition or pacemaker.